Can I use the Vitaport with Stellate Harmony?
Yes, you can use the Vitaport recorder together with Stellate Harmony. You must set-up the recorder as usual, using the Columbus program.
As you know, you can set the storage rate of each channel of the Vitaport recorder independently. Because of the restrictions within Stellate Harmony, the storage rate difference between the minimal used rate and maximal used rate may not exceed a factor of 64.
After the recording is done, you must convert the RAW-data file from the recorder to the EDF-format using the Columbus program. Stellate Harmony can import this data.